By creating those folders, the proplem seems fixed. Ideally the output file would be formatted something like: Page gives the rate commands. Anyone know why I can’t get any data from it? Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
|Date Added:||27 December 2011|
|File Size:||62.80 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
It would be helpful if someone can help us resolve this. I have included a screenshot of this, and the for loop is just to automatically generate the names for each channel.
In order to resolve this issue, a “Decimate 1D Array” function is used in order to separate values from individual channels into separate rows or columns in an array. Page gives the rate commands. Message 9 of This is very odd behavior.
Seems you are also confused about the keithlwy Apart from the Function 0 i.
keithley driver problems – Discussion Forums – National Instruments
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Ensure you can talk using the diagnostic tool Anyone know why I can’t get any data from it? How often does it occur and what steps do you take to reproduce this? It seems for me than you could read about producer – consumer: It’s the same problem as described in here: By creating those folders, the proplem seems fixed. Message 6 of You should see keitlhey least wo entries if all is working: I couldn’t find a manual for thebut I found one for the Keithley The Keithley has a configurable scan rate of Fast, Medium, and Slow.
Click on the Read tab and press execute Are you successfully communicating with the instrument using the Visa Interactive test mentioned above?
Message 24 of I figured the problem is in the converter. Hi DaleYes ,its very odd for me too. Once you have determined the model, ensure you have installed the correct and latest driver. And is there a way to get the USB show up?
Keithley 27XX Switch,Meter,Data Acquisition System
Message 25 of Thanks for the VI. I have the exact same problemi am using same program.
Hi Jesse, I couldn’t find a manual for thebut I found one for the Keithley Message 4 of 5. Thanks for the info, will try that! I also had luck changing the aperature timing by placing it after the config measurement.
Continuously Write to Spreadsheet Keithley 27XX – Discussion Forums – National Instruments
In this moment I am using a. Additional Images or Video. I’ve been struggling with this for a while now. I tried using it for the measurement of resistance but seems like it gives an error.